Posted by Jerry Knisley, Last modified byon 09 December 2014 12:25 PM
Driver providing 2-channel sidescan information from analog input. A separate program (Ascan.exe) is run automatically to sample the ping trigger and digitize the analog inputs. The digitized data are passed to HYSWEEP Survey through shared memory.
The is no setup form in HYSWEEP Hardware. However, clicking “Analog Sidescan Monitor” (on the taskbar) during HYSWEEP Survey will show the form for sidescan configuration on the fly. There is an image attached at the bottom to show what this looks like.
Input: Select one of the voltage range options.
Trigger: Ping is on low to high or high to low transition of the trigger signal.
Gain: Gain constant (voltage multiplier).
Threshold: This is the A/D count level at which trigging occurs.
Test: Simulate data provides a dummy amplitude time series for each channel.
Connected via a National Instruments multifunction I/O device – PCI board or PCMCIA card. The device is plug and play – no configuration required.
Offsets – Sonar Heads 1 and 2:
See Appendix A: Analog Sidescan Interfacing.